Message 5 of Most Active Software Boards: Thank you so much for your help!! Message 15 of Message 3 of One of my other approaches was it, that the interconnected adapter disturbs the IDN – transmission, or even brings in it’s own IDN. The BK precision one has its own converter and for the other 2 I use this converter:.
|Date Added:||14 October 2007|
|File Size:||6.40 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Instrument driver for Hewlett Packard EA – Discussion Forums – National Instruments
Thank you very much for your help, Keksbold. Is it an NI device or 3rd party? I somehow think it would be a bad idea to just disable all the stuffs and hope fopr the best I’m trying to program the power supply to sweep its current range and output each value.
Message 9 of Thanks for your reply! To be sure I had things working correctly, I first connected the power supply to an “on s3631a comport and used Tera Term to verify I had the correct cable, baud rate and such.
Most Active Software Boards: How could I check? They both don’t work and as they are quite new, I am pretty sure they are not broken.
Agilent E363X Series Power Supply
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Try looking at the manual on the agilent website http: Oops I made a mistake above.
This labviw to work, but the next error appeared when it tried to reset. I was planning to use the power supply in a dynamically controlled system, but unfortunately it always takes a few seconds to change the voltage level.
Labview -> HP EA PowerSupply Problems – Discussion Forums – National Instruments
Did you say that you had it communicating once that way? Message 2 of You must have a different e3613a of drivers than I for the E Also connecting to a comport only means you are able to get as far as the com port itself.
Message 9 of I got my drivers from the NI website. If you jump to sending another command after 1 the instrument can get confused. Got the Agilents working with a null modem inbetween.
There are also default com port settings in the Device Manager of Windows. The strange thing is that when you change this, the DC Power Supply changes correctly, follows these settings.
Message 3 of The BK precision one has its own converter and labviiew the other 2 I use this converter: Actually I have to admit that I am just using a gender changer. After that I had to remove the error queries from the Configure Output and the Configure Current Limit as well, which made these work too.
Do you have the current limit set to s3631a Most Active Software Boards: Message 15 of I’m not sure whether the current limit is set to 0. That made the initialize sub VI run. Message 7 of This time you need to make sure the output side which did not connect anything.